This item is no longer available. Suggested replacements. UVAS sc Probe, 1 mm, with SC Controller · Get Quote. Continuous UV Results 1 – 20 of Search results for sc Amtax sc Ammonium Analyzer, mg/L NH₄-N, V, 5 m Filterprobe sc. This item is no longer available. Suggested replacements. UVAS sc Sensor, 5 mm, with SC Controller · Get Quote. Continuously protect plant treatment.
|Published (Last):||9 February 2017|
|PDF File Size:||11.11 Mb|
|ePub File Size:||2.63 Mb|
|Price:||Free* [*Free Regsitration Required]|
Don’t hac me this message again. Don’t have an ahch Setting Up System Security Operation 4. Section 7 Troubleshooting 7. Pull the lamp leads and connector through the hole in the lamp housing. Measurement Xfer Transfer 4. Index Index Numerics 90 Degree Detector Controller Display Features Operation 4. Instrument Setup Operation 4. Measured reading based on a gain of 1. The frequency will depend on the nature and concentration of dissolved and suspended solids in the sample.
In addition, the passcode also regulates function selection for relay options. Section 1 Specifications Section 1 Specifications Specifications are subject to change without notice.
Page 71 Index Sample Line Installation Refer to Figure 29 and perform the following steps to replace the fuses: Hacy pressing ‘print’ button you will print only current page.
Rinse with deionized water before calibrating. Page 20 Installation 3.
Section 11 Limited Warranty In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase price, subject to the pro-rated schedule above, excluding original shipping and handling charges.
Index Numerics 90 Degree Detector After all plumbing and electrical connections have been completed and checked, place the head on the body and supply power to the system.
Section 4 Operation Section 4 Operation 4. The passcode is factory set to sc the five digits must be followed by a space to remove the trailing asterisk.
Hach-Lange SC User Manual | 58 pages
Wire gauge less than 18 AWG is not recommended. Table of Contents Add to my manuals Add. Page 50 Maintenance a.
Sc10 display will flash on startup, when the hold outputs function has been activated, and when the filter function signal average is changed to a different value. Section 7 Troubleshooting Section 7 Troubleshooting 7.
Wet Verification System Startup 5. Dry Verification System Startup Two types of verifications are offered.
Hach-Lange SC 100 User Manual
The data logs store the measurement data at selected intervals. Sensor Setup Menu Operation 4. Do not touch the new bulb with bare hands. Enter text from picture: Got it, continue to print.
Changing The Sensor Name Operation 4. To ensure that the protection provided by this equipment is not impaired, do not hacj or install this equipment in any manner other than that specified in this manual.
Mounting The Controller Installation 3. To locate the representative nearest you, send e-mail to intl hach. Hah contact the Hach Service Center serving your location.
Remove the quick connect fitting and wires and install the threaded plug on the opening to maintain the environmental rating.